ICV. Vacuum-Line Cleaning Made Simple.Many vacuum-line cleaning methods are messy and take time. ICV integrates right into your dental cabinet and cleans vacuum lines quickly and efficiently.
Simple 3-Step Operation
Just connect the HVE and saliva ejector to the ICV, open the valve, and press the start button. In two minutes or less the vacuum lines are purged.
Safe, Easy-Carry Tank Design
The tank design with easy-carry handle and secure lid eliminates the potential for chemical spills when transporting the tank between fill ups.
Two minutes per cycle. Cleans vacuum lines quickly and efficiently.
Integrates right into your cabinetry. Provides you with easy access to ICV.
Easy-to-fill tank. Allows you to safely add cleaner and water.
Compatible with most cabinets. Requires minimum cabinet module dimensions for fluid tank storage 7" W x 20" D x 21" H (178 mm x 508 mm x 533 mm).
ICV Vacuum-Line Cleaning SystemSee how to use the ICV vacuum-line cleaning system in A-dec Inspire dental cabinets.
Safety Data Sheets (SDS)
What is the reservoir capacity?
The ICV tank holds 8 liters or 2.1 gallons.
How do I know when the reservoir is low?
There is a low fluid indicator light on the ICV face plate that indicates when the tank is low.
What if I only have 1 vacuum handpiece?
Each handpiece port on the ICV face panel has a check valve that allows the cleaning solution to only flow to the ports that have an active vacuum handpiece attached.
Do I still need to change my dental unit’s vacuum solids separator and canister screen?
Yes. Continue to follow the maintenance protocol of the dental unit’s vacuum solids separator and canister screen.
How do I adjust the ICV timing interval?
The A-dec Inspire ICV timing can be adjusted from the A-dec Inspire control center settings. See the A-dec Inspire Dental Furniture Instructions for Use (page 9). The Preference Collection ICV requires a service technician to adjust the timing.
How much fluid is used per activation?
The fluid used per activation is based on the ICV timer setting and the performance of the facility’s vacuum pump system.
How much fluid should be used per activation?
Consult the vacuum pump directions for use of the recommended cleaning product, typically they recommend 1 liter or 1 quart.